SIMCHECK II Line
SIMCHECK II LT PLUS
Refurbished LT PLUS
SIMCHECK II Adapters
Why Test Memory?
How to Purchase
Service & Upgrades
Contact us at
- Q. If necessary, can I replace the RAMCHECK
Power Supply with one that we have in house, or with one acquired
A. We strongly recommend that replacement power supplies be acquired from us,
as we do fine tuning of lower voltage to reduce the heat that is dissipated
inside the tester. However, if you have technical knowledge of power supplies,
you may replace it with a new one, only if it is polarized in the same manner
as the original Power Supply (center positive) AND it meets the output rating
for CURRENT and VOLTAGE. For RAMCHECK the power supply's voltage must NOT
exceed 6.90V at actual current of 3Amp. We use a 7.00V/4.4A power supply which
is carefully adjusted for 6.90V at 4Amp.
- Q. Can the RAMCHECK power supply be
used with SIMCHECK II and vice versa?
A. NO. RAMCHECK requires slightly lower voltage, as detailed above, in order to
reduce the heat inside the unit. During factory upgrade from SIMCHECK II to
RAMCHECK, our technicians carefully adjust the original power supply for
RAMCHECK levels, or completely replace it.
If you have additional questions, please feel
free to email us at email@example.com.
Please remember to include your phone and fax numbers.